In order to solve the problem of low sampling rate and to meet the requirements of high frequency swing mirror measurement, based on the principle of equivalent sampling, a multi time series high accuracy non-contact swing mirror detection system is designed. First, with the laser pulse control circuit, multi time sequence flashing laser lighting is achieved and the spot position is obtained on the CCD target plane. Then, according to the image of the laser spot position, calculate the mirror swing angle and angular velocity by the time interval between adjacent spot positions. The experimental results show that the angle resolution of the detection system is 0.005°. The time resolution can reach the microsecond order, and the angular velocity measurement error is less than ±7%. The system improves the sampling frequency of swing mirror measuring and meets the requirements of high frequency swing mirror test.
Multi time sequence flashing laser test for high frequency swing mirror
First published at:Jul 01, 2018
1 Xu B Z, Liu T G, Wang M, et al. Research on laser marking technology for free-form surface[J]. Chinese Journal of Lasers, 2010, 37(8): 2165–2169.
2 Zhao H X, Zhao J K, Tian L D, et al. Detection of angular tracing precision of rotating mirror house in streak camera[J]. Optics and Precision Engineering, 2015, 23(12): 3303–3308.
3 Ravindra Singh R, Mudgil A, Prakash C, et al. Precision gimballed mirror control in remote sensing LIDAR for environmental monitoring[J]. Proceedings of SPIE, 2006, 6409: 64091F. DOI:10.1117/12.694397
4 Huang Z H, Zhang C, Cai H Y, et al. The research of joint control algorithm of moving target dome projection based on two-axis turntable[J]. Opto-Electronic Engineering, 2014, 41(9): 32–37.
5 Pu Z B, Tao W, Zhang Z. Angle measurement with optical methods[J]. Optical Technique, 2002, 28(2): 168–171.
6 Filatov Y V, Loukianov D P, Probst R. Dynamic angle measurement by means of a ring laser[J]. Metrologia, 1997, 34(4): 343–351. DOI:10.1088/0026-1394/34/4/7
7 Ikram M, Hussain G. Michelson interferometer for precision angle measurement[J]. Applied Optics, 1999, 38(1): 113–120. DOI:10.1364/AO.38.000113
8 Wang W Q, Liu G D, Pu Z B, et al. Measurement of two-dimensional small angle by means of matrix CCD[J]. Semiconductor Optoelectronics, 2004, 25(2): 134–135, 138.
9 Shi Y L, Gao Y G, Zhang L, et al. Improvement of measuring accuracy of CCD laser autocollimator by changing hardware parameters[J]. Optics and Precision Engineering, 2008, 16(4): 726–732.
10 Zhang G Y, Jiang H L, Xu X P, et al. Laser dynamic testing method for swing angle of scanning mirror of infrared earth sensor[J]. Acta Optica Sinica, 2007, 27(5): 875–881.
11 Wang Y, Yan C X, Hu C H, et al. Non-contact rotation angle measurement system for scanning mirrors[J]. Optics and Precision Engineering, 2014, 22(4): 850–855.
12 Qiu D Y, Tian S L, Ye P, et al. Research and implementation of a random equivalent time sampling based on parallel structure[J]. Chinese Journal of Scientific Instrument, 2014, 35(7): 1669–1675.
13 Jin Z L. Error analysis of centroid algorithm based on CCD[J]. Spacecraft Recovery & Remote Sensing, 2011, 32(1): 38–44.
Get Citation: Cai Huaiyu, Liang Zhimin, Huang Zhanhua, et al. Multi time sequence flashing laser test for high frequency swing mirror[J]. Opto-Electronic Engineering, 2018, 45(7): 180052.